Open as PDF
94 MGP D5 Dicom Theater Lite
3. Click the OK button.
The calibration and the luminance will be reset to the latest
When you start up the system, the calibration is also reset to
the latest calibrated position. The luminance, however, is
kept at the latest value.
To view a graph showing the result of the latest calibration:
1. In the Calibration tab dialog, click on Show calibration
This button is available only if the display has been
calibrated with a sensor before.
2. A graph is shown displaying the latest calibration compared
to the theoretical display function.
To revert to the default factory settings:
1. In the Calibration tab dialog, click on Reset.
2. As a result, the settings are restored to the default factory